Stroboscopic 100 to 10,000 FPM/RPM
Accuracy 100 to 5,000 FPM/RPM)± 1 digit.
over 5000 FPM/RPM – ± 0.05%
Resolution Less than 10,000 FPM/RPM
Over 10,000 FPM/RPM -10FPM/RPM
Sampling Time 1 second.
FLASH TUBE SPECIFICATIONS
Flash tube Xenon lamp
Flash Duration Approximately 60 to 1,000 microseconds.
Flash color Xenon white 6,500 K degree.
Operating duty For prolong life and safe operation, please adhere to the following duty cycle:
Cycle Below 1,000 RPM – 30 Minutes. Above 3,600 RPM – 5 Minutes. Always allow a 10 minute cooling off period between cycles. Flash tube when the instrument start to flash irregularly at speeds of 3600 RPM/FPM or more.
Preparation (a) Plug unit into a properly grounded 110V AC, 220V AC or 230V AC outlet.
(b) Turn the power switch to “on” position.
(c) Determine the range switch to “Low” or “High” position.
Checking Speed When checking speed, care must be taken to insure that the strobe is flashing in unison (one to one) with the object being monitored. A Stroboscope will also stop motion at 2:1, 3:1, 4:1 et., this is normally referred to as harmonies. To besure of unison, turn the dial until two images appear – this will double the actualspeed. Then lower the flashing rate until a single and stationary image appears -this is the actual true speed.
Checking Motion For motion analysis, simply locate the actual speed as mentioned above and then
turn the dial slowly up or down. This will give a slow motion effect allowingcomplete inspection.
Accessories included ( termasuk )
Power Cord 220 Volt